ROV Coordinate Estimation With 0.07% Slant Range Accuracy18/01/2012
|The Evologics S2C USBL (Ultra-Short Baseline) underwater positioning system demonstrated a consistent performance in a recent series of pool tests, estimating the position of an ROV for baffle bolt inspection in nuclear power plants. The system estimated the distance to the ROV resulting in a RMS deviation of 0.04° at 2.1m distance. This result corresponds to a 0.07% 1 drms slant range accuracy.|
The Submarine System For Inspection (SUSI) ROV, developed by AREVA, is a device used for ultasonic inspection of nuclear reactor vessel baffle bolts. The ROV's positioning system is required to document time and exact location of each measurement taken as the ROV examines numerous bolts that hold the baffle plates together. Such positioning must deliver an accurate performance without obstructing or slowing down the inspection procedure.
In December 2011, the Evologics USBL system was tested in a tracking scenario that emulates the testing procedure. These trials were conducted in a test pool, as such a positioning system must deliver accurate results in a confined space, where noise and multipath challenge the performance of underwater acoustic devices.
An S2C USBL device with a conical transducer beam pattern served as a transceiver and was mounted at the side of the test pool. An S2C underwater acoustic modem, mounted on the ROV with the transucer pointing vertically upwards, was used as a transponder. The ROV was located at the bottom of the pool within the transceiver's transducer beam.
To verify the results, the ROV was moved across a vertical 85x100mm testing grid with reference points that represented the actual baffle bolts. The system demonstrated consistent results and the 0.07% slant range accuracy was confirmed.
comments powered by Disqus
Read more about:
Tracking ROV USBL Positioning vessel
Supplier: EvoLogics GmbH
More news from this supplier:
Dead Sea Underwater Freshwater Springs Surveyed with USV
Acoustic Link for Real-time Tsunami Monitoring System in Japan
SUNRISE Project Connects AUV Swarms into Internet of Things
EvoLogics to Unveil a Series of Compact Underwater Acoustic Modems
Upgraded Developer Edition Modems UnetStack-compatible
Running Virtual Underwater Modems
Fleet Of Autonomous Vehicles Connected Into Acoustic Network
Collaborative Underwater Communication Projects
Latest EvoLogics Developments at Ocean Business
Successful Beta-testing for Underwater Modem Emulator
Surveyor Interceptor ROV First Commercial Work for Gassco
Michel Huet Knight of the French Order of Maritime Merit
Port of London Gets SONIC 2024
DOF Subsea Supports ROV MATE Competition
Swirling Currents Deliver Phytoplankton Carbon to Ocean Depths
Hydromagic v7.0 Released
Eight Galileo Satellites in Orbit
IMCA Paper Basis For Updating DP Guidelines
OCEANS 15 Washington Opportunities Related to Change
nke NOSS In Situ Density Sensor Presentation
Teledyne BlueView 3D Scanning Sonar in Use for Inspections
Teledyne BlueView 3D scanning sonar is used in infrastructure inspection such as post scour remediation inspection. Infrastructure managers verify construction was completed to contract specifications and for pre-construction inspection to assist with project planning.